The Corus PTZ is a key element in the whole Actaris chain, from the meter to the billing data, because it converts the actual volume measured. For each FLUXI TZ and DELTA meters, correction according to pressure, temperature and compressibility factor (meaning the transformation in real. CORUS is an electronic volume converter dedicated to commercial and industrial applications. It converts the actual volume measured by the gas meter to.
|Published (Last):||17 March 2009|
|PDF File Size:||16.45 Mb|
|ePub File Size:||18.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
The presence of an optical head is magnetically detected.
Solenvis SL79 Itron-Corus gas volume converter – Solenvis Flowmeters
For safety reasons, it is strictly forbidden to use other modules than those specified by ‘Actaris’. When CORUS is codus in the external power supply mode, the supply has to be provided through an intrinsically safe module with electrical compatible parameters ; Use of any other type of module is strictly forbidden. Nature of the event? The RS port is by default in a power-down mode.
VERIBOX-MINI – Automatic meter Reading / AMR – ATEX Datalogger
All cable glands have moreover to be tightened enough. Battery Current value Mode: CORUS uses the measured working values of volume, pressure and temperature to provide: The converter must then be actrais into the ‘external power supply’ mode. A hinge on the left of the box makes any operation on the field very easy, providing a clear access to all major parts of the product terminals, switches, internal battery,… Corus PTZ is built with three electronic boards: Program the required parameters Keyboard or communication?
CORUS uses the measured working values of volume, pressure and temperature to provide:. Following the installation procedure, Corus PTZ has to be carefully closed using the four screws placed on each corner. All daily consumptions are stamped from the gas hour to the gas hour of the next day. The activation duration of the alarm outputs can be selected.
Itron Corus Gas Volume Converter
During the main cut, the internal backup battery will ensure the total functioning of Corus PTZ and the battery counter will be decreased with the same management described in the “battery alarm” chapter. Any attempt to open the box lets permanent visible marks on the seal or on the enclosure.
Select the type of alarm to be resetted: These plates are screwed and sealed.
Then, It is strictly forbidden to replace the battery unit by any battery other than the one specified by ‘Actaris’ reference: Its volume input is a low frequency one that can be connected to any Ex compliant gas meter that provides such information.
Contact us to get a quote or a custom-tailored demonstration with one of our business advisors. This input shares the same terminals J3 and J4 and the same cable as the main LF input. Then, this alarm is set as long as the external supply is not present. The actari is then displayed. It must be preferentially screwed to the “Pr” reference pressure tap of the meter: The alarm outputs can remain active: The system is built in an IP65 enclosure, for wall or meter mounting.
The ‘OK’ displays the date and time, as well as the firmware version numbers.
This alarm is not metrological and is dedicated to the control of the whole metering chain. All monthly consumptions are stamped from the gas day at the gas hour to the gas day of the next month at the gas hour. A press on the OK key displays a chart for the selected value. They can be either: If the function is activated, an alarm of the above list is set if the corresponding flow-rate gets higher or lower than the threshold.
The user can program a number of pulses Np and a duration Dp. The tamper input can be set into one of the following modes: We would like to thank all our customers for your continued support over the last 5 years.
Based on the use of flash memory, its architecture allows the download of a new firmware release through a laptop without board modification. Recorded data and number of records can be freely set. Bank of England select Solenvis to supply clamp-on metering.
Installation and program procedures must be carried out and inspected by qualified personnel. The five keys allows to move into the various menus as described pts.
The terminal J6 that correspond to the serial port is located in the customer area: Solenvis carry out an extensive meter upgrade project at Oxford University. The closing time of the volume pulse outputs can also be programmed default value is ms.
The input pulse weight can be chosen among the following values: Input pulse weight, unconverted Index, Converted Index: